(1) The electric heating incubator should be handled by full-time personnel. No matter how the power switch and the adjusting twist in the actual operating plate are fixed for one day, it is not necessary to shake at will in order to prevent harming the temperature, CO2 and humidity level in the incubator, and to reduce the sensitivity of the equipment.
(2) The water registered for participation should be pure water or non-Zhaizi water as far as possible, so as to prevent the erosion effect of mineral storage accumulating in small boxes filled with water. Change the water as much as possible over the years. Check regularly whether the water in the seizure box is enough.
(3) The internal policy of the box should be regularly scrubbed and disinfected with disinfectant_shelf can be extracted, cleaned and disinfected to avoid other micro-living substances pollution, resulting in experimental failure.
(4) Search and seize the overheating insurance device on time to prevent overheating. The method is to check the alarm by pressing, rotating and fixing the fixed screw until the overheating alarm system sounds, and then turn off the overheating safety lamp.
(5) If the carbonate gas is not used for a long time, the CO2 switch should be turned off to avoid the failure of the CO2 regulator.
(6) The CO2 used is as pure as possible, otherwise the sensitivity of the CO2 controller and the environmental pollution of the CO2 leaching equipment will be reduced.
(7) To maintain the stability of CO2 in the incubator, a water-holding incubator should be placed at the bottom of the incubator.